Our innoCon 6800CL residual chlorine analyzer is based on a double platinum potentiostatic method.
The measurement and the reference electrode maintain a stable potential, and different measured components will produce different current intensities at that potential. In the measurement process, chlorine molecules or hypochlorite will be consumed, as the current intensity produced is related to the concentration of residual chlorine in water.
When pH value is In the range of 6-8, the measured signal decreases while the pH value increases, so
innoCon 6800CL chlorine can only be used in the condition of pH ≤ 8. If the pH value in the water is constant,
the effect of pH can be compensated by calibration on site.
The same potentiostatic method is used to measure ClO2 and O3.
● No need colorants
● No need to replace the membrane and electrolyte gel
● Easy to install
● Low maintenance costs
● Zero-point stability
● Flow Cell to keep the flow constant
Measuring range:0 - 2.000, 0 - 20.00 ppm (mg/L)
Resolution: 0.001 / 0.01 ppm (mg/L)
6 - 8 pH (Cl2)
6 - 9 pH (ClO2, O3)
0 - 60 ℃(Working temperature)
1 bar (Max. pressure)
10 - 30 litre/hour, constant (Flow)
Dimensions: 120 * 12mm
|innoSens 710||1||Thu, 10 Oct 2019 09:33:04 GMT|
|innoCon 6800CL datasheets_EN||1||Mon, 19 Mar 2018 10:01:20 GMT|
|35-0710-00||innoSens 710 Potentiostatic sensor|
|33-6801-40||innoCon 6800CL Chlorine / Chlorine Dioxide / Ozone controller, 90 - 260VAC|
|50-0711-00||Flow cell for innoSens 710|
Sensors assures precise measure in analysis
High quality provides high performance and simple operation
Human design of analyzers manage long lifespan